Fig. 1 My scr-triac test setup.
Simple Triac-SCR Test Lab for You Tube
by Lewis Loflin
This page is related to three You Tube Videos on simple test circuits for SCRs-thyristers and Triacs. Most of the explanations are in the videos.The first part of a three-part lab on testing SCRs and Triacs. http://www.bristolwatch.com/ccs/scr_triac_lab.htm
- You Tube Videos
- How to Check SCRs and Triacs - Simple Circuit Part 1
- Testing SCRs Simple Circuit Part 2
- Testing Triacs Simple Circuit Part 3
This problem came up when my Kuman and MK-168 transistor checkers would test most SCRs unless they had very sensitive gates, and no triacs would test at all. At least those I had shown to the right in Fig. 1.
The test assembly used a ZIF socket connected to the home built proto-board in the upper center of Fig. 1. A 25.2 volt A.C. transformer is outside the photo.
A 24-volt lamp is used as a load and measurements are made with a Cen-Tech DVM. A white socket in the center of the board allows one to plug in a TO-220 style SCR or Triac or an external adaptor cable with alligator clips for oddball parts that won't fit it or the ZIF socket.
Fig. 2 is the electrical schematic for the test board. Power is supplied by the transformer and gate current for either device is controlled by R1 a 5K potentiometer. A diode is used for SCR tests and is jumpered for triac tests.
Fig 3 illustrates the connection for testing SCRs S1 is open. When the SCR is fully turned on with R1 is acts as a half-wave rectifier and the DVM will read ~12.4VDC based on the 28VAC from my particular power transformer.
Notice that the gate turn voltage varies from one SCR to the next. Two were very sensitive while some required much greater turn on current for full output voltage to the DVM.
Fig 4 illustrates an SCR being used as a half-wave rectifier with voltage calculations.
Fig 5 uses a potentiometer to vary the trigger point on the half-wave. The formula in the slide only works if the SCR is fully on.
Fig. 6 illustrates a triac connected to the test setup. When the triac is fully turned on by R1 the DVM reads 0 volts DC and when on the AC setting reads 27VAC. (One volt is across the test triac.)
Be sure S1 is closed!
As the attached video noted when R1 was turned up one side of the triac switched on and the device acted like an SCR producing a DC voltage. As I continued to adjust R1 the other side also turned on. The lamp became bright, no DC voltage just AC across the lamp.
The answer to the problem is Fig. 7. A triac acts like two back-to-back SCRs with the gates connected in common. Each "SCR" has a different cut on voltage which is why one turned on and acted like an SCR until the 2nd SCR turned on.
Fig. 8 our solution to the tun on problem. Let's argue (Fig. 7) Q1 turns on at 22 volts and Q2 at 28 volts. The diac that was inserted into the gate circuit fires at about 30 volts dumping sufficient current into both sides at once.
That switched on Q1 and Q2 together regardless of differing gate turn on voltages.
- Experiments with TL431 Shunt Regulator
- TL431 Precision Current Regulator Circuits
- TL431 Based Current Limiter Constant Current Source Circuits
- TL431 Shunt Regulator Circuits
- Comparing ATmega328 Based Transistor Checkers
- Basic MOSFET Transistor Test Circuits
- High Voltage MOSFET Switching Circuits
- Quick navigation of home page:
- Arduino Microcontroller Projects
- General Electronics Learning and Projects
- Raspberry Pi and Linux
- Connecting a PC Printer Port to Electronics with Python
- Microchip PIC 18F2550
- PICAXE Microcontroller
- LM317 Adjustable Voltage current Boost Power Supply
- LM317 High Power Constant Current Source Circuit
- Constant Current Circuits with the LM334
- LM334 Constant Current Source with Resistive Sensors
- LM317 Constant Current Source Circuits
- Introduction Hall Effect Switches, Sensors, and Circuits
- Using Ratiometric Hall Effect Sensors
- Pulse Width Modulation Power Control for Microcontrollers
- Introduction to PIC12F683 Programming
- Basic Transistor Driver Circuits for Micro-Controllers
- Opto-Isolated Transistor Drivers for Micro-Controllers
- ULN2003A Darlington Transistor Array with Circuit Examples
- Tutorial Using TIP120 and TIP125 Power Darlington Transistors
- Driving 2N3055-MJ2955 Power Transistors with Darlington Transistors
- Understanding Bipolar Transistor Switches
- N-Channel Power MOSFET Switching Tutorial
- P-Channel Power MOSFET Switch Tutorial
- H-Bridge Motor Control with Power MOSFETS
- More Power MOSFET H-Bridge Circuit Examples
- Build a High Power Transistor H-Bridge Motor Control
- Arduino Nano Test Template
- Arduino Solid State Relay Motor Enable Control
- Arduino Blink LED Tutorial
- Arduino SSR Power Enable Program
- SSR Based High Voltage H-Bridge
- Arduino H-Bridge Motor Control Program with LCD Display
- Arduino XOR Blinks LED
- Arduino IF Statement Code Examples
- Arduino Motor Control Program Using IF
- LM317 Adjustable Voltage Source Current Boost
- Geiger Counter Basics Radioactivity
- Introduction to Geiger-Mueller Counters and Electronics
- Astable CD4047 Geiger Counter Power Supply
- CD4047 Monostable Multivibrator Circuit
- Getting Real About Radiation Myths and Hazards
- Uranium Hype-Facts and Virginia Uranium
- Uranium Basics and Isotopes
- Climate Change, Climate Scams, and Volcanoes
- Technology is Why the Jobs are Not Coming Back
- Off-the-Shelf Technology for Space Exploration